Indicator for instrument when power on. Other kind of paper may not render a clear permanent trace and may damage the printing mechanism. Please consult distributor or manufacture for detail of how to purchase the paper. V2: Fourth inter-costal space at left border of sternum.
|Genre:||Health and Food|
|Published (Last):||18 March 2011|
|PDF File Size:||6.12 Mb|
|ePub File Size:||10.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Our company has the sales record and customer archives for each instrument. The remark includes: lead sign, sensitivity, paper speed, filter state. The sketch map and components name Front view Side view Page 10 User Manual Function button: system menu Function button: upwards Function button: downwards Function button: leftwards Function button: rightwards C.
Indicator Definition The indicator turns green when there is AC power supply, and when the indicator turns green and red same time it is being recharged.
Indicator for instrument when power on. V2: Fourth inter-costal space at left border of sternum.. Page 16 User Manual unit.
Page 17 User Manual Page Chapter 11 Grounding And Power Connection User Manual Chapter 11 Grounding and Power Connection Make sure the status of the instrument is power off, and then make the instrument be properly grounded through a 3-prong outlet.
When the outlet, a grounding cable may be utilized to connect the grounding terminal of the instrument. This battery is with perfect automatic charge and discharge monitoring system. When you connect power supply adapter with alternating current, the charge will be start automatically.
When this device be open, an icon be displayed on top right corner of LCD screen. Page 21 User Manual placed well. Stop printing and print date again after collecting the wave. When the leads state is "DROP", leads shown on the screen have been off. Please reconnect them. Page 22 User Manual Press to enter above interface, you can choose relevant item by press , then you can press to adjust the content, after setup, press to be back.
Turn off is for protecting circuit. Is the electrode and skin covered with enough Gel? Any maintenance or update should execute by the trained and authorized professionals from our company. The maintenance should be done with the original accessories from our company. Page 31 User Manual Explanations of symbols on unit Symbol for "applied parts" the electrodes are type CF applied parts. Symbol for "environment protection" - waste electrical products should not be disposed of with household waste.
Please recycle where facilities exist. Check with your local Authority or retailer for recycling advice.
Electrocardiógrafo Contec Ecg 300G 3 Canales Digital
Contec ECG 100G User Manual
Contec 300G User Manual